Why would the pressure transmitters be changed from Bar(G) to Bar(A) ones? The pressure in the flow measuring system has dropped from 25 Bar(G) to 5 Bar(G). Flow is worked out using ISO 5167.
|Thread starter||Similar threads||Forum||Replies||Date|
|Help with Changing ID and Baud on SDM120, EM23 and EM24||Modbus||1|
|A||Honeywell DCS - PID Time based Setpoints Changing||General Automation Chat||1|
|I||Changing Switch Configuration||Power Generation||0|
|E||Changing ARCNET Card||Power Generation||3|
|U||Changing a Modbus Address||Modbus||1|