How I can adjust the temperature in a Honeywell UDC1200 and UDC 1000? When I simulate 1500 degrees it shows me 2 degrees below tolerance.
2 Deg for a 1500 Deg span translates into
+/- 0.13% based on Max Range, or +/- 0.26% Mid-range basis
typically, depending on your sensor you find that in the actual operating plants a mid-range error is +/- 6 to 7% on a good day,
you can get greater precision with national lab traceability, under lab conditions with special design platinum resistance sensors. In an operating plant, day after day, at temperatures >400C that is another matter...